Solar Light’s Model XPS-300 Power Supply is a highly stable current source for xenon short arc lamps which is specifically designed to work with Solar Light’s 601-Series and 16S-Series Solar Simulator models, with universal mains operation from 90-250VAC.
The rear panel has connection points for lamp power, chassis safety ground, 24VDC fan power, a normally closed fault loop circuit, 24VDC solenoid driver and signal lines indicating shutter and filter operating positions.
The XPS-300 is power regulated and power limited to simplify operation. A single front panel knob determines the power setting for the lamp, and a connector supports PMA/DCS meter options and access to 24VDC shutter control. The internal ignition counter stops the automatic ignition cycle after approximately 5 attempts.
An external fault loop is used to provide thermal shutdown in the event of a lamp overheating or the fan failing. The fault loop circuit provides an intrinsically safe, normally closed circuit for remote sensing elements such as interlock switches, thermal switches or vacuum sensors. The system will not restart until the lamp enable switch is reset.
The power supply senses a lamp’s end of life condition and will shut down when the lamp voltage exceeds 30VDC. Output is short circuit protected and “arc to ground” protected. The power supply chassis is internally protected with a thermal shutdown feature if the internal temperature exceeds 90°C.
1 YEAR STANDARD WARRANTY
SOLAR LIGHT COMPANY warrants that the equipment has been carefully tested, inspected and left the Factory in proper working condition, free of visible defects. SOLAR LIGHT COMPANY warrants the equipment to be free from defects in material and workmanship, under normal use and operation, for a period of one (1) year after the date of initial operation, or demonstration; or eighteen (18) months from the date of shipment from the Factory; or whichever occurs first. The warranty is limited to the free replacement of all-defective parts and free labor during the first year of warranty. Defective parts replaced under this warranty shall become the property of SOLAR LIGHT COMPANY. The above stated Warranty is subject to the following conditions: Equipment being installed within the United States of America, Canada, or sold by authorized SOLAR LIGHT COMPANY representatives or their agents outside the continental United States. The equipment being installed is operated and maintained in accordance with SOLAR LIGHT COMPANY’s operating instructions, maintenance instructions and specifications. Utilization of only SOLAR LIGHT COMPANY’s authorized parts and components. The utilization of a duly authorized representative for all work performed on SOLAR LIGHT COMPANY equipment during the warranty period. Any variation(s) to the above warranty must be in writing and mutually acceptable by SOLAR LIGHT COMPANY and the customer. The warranty shall not apply to damages resulting from errors in installation, nor shall it apply to any equipment, which has been subject to damages, alterations or misuse by the purchaser. Parts of the equipment, which would be generally considered as expendable service items during normal use, such as fuses, lamps, detector shields, etc., are not covered under the warranty. THERE ARE NO WARRANTIES, WHICH EXTEND BEYOND THIS DESCRIPTION, AND EXPRESSLY NO IMPLIED WARRANTY OR MERCHANTABILITY OR FITNESS FOR A PARTICULAR PURPOSE. IN NO EVENT SHALL SOLAR LIGHT COMPANY BE LIABLE FOR SPECIAL, INCIDENTAL, OR CONSEQUENTIAL DAMAGES OR ANTICIPATED LOSS OF PROFITS SUFFERED BY THE PURCHASER. Extended Warranty Plans are also available.